13400 - HPL 1.6 - high power laser
HPL stands for High Power Laser and characterises the new equipment for lasertherapy, which are provided with the new high power infrared diodes with a great depth of action.
In order to activate the natural repairing processes it’s advisable to involve the different cells with an energy density able to cause a biological effect. The power diodes in the infrared field, used for the HPL, assure such a depth of action and power to involve the different cells and activate repairing, analgesic and anti-inflammatory processes.
The suitable parameters for treatment can easily be obtained: when the therapeutic target is established, the operator can act indifferently on application time or energy density, deciding at the same time to keep unchanged one and modify the other. The computerised system computes automatically the missing parameter in relation with the delimited scanning area.
The emission mode is both continuous and pulsed. In continuous mode the power of laser is always the highest and it is suitable for treating wide areas of the body, for which it is necessary to deliver a high energy density in relatively short times. The pulsed emission instead is used to avoid an early saturation to laser energy of the treated area.
The HPL scanning plane has an automatic rotation system from +90° up to -45°, and allows treatment in a seated, upright of lying position.
Two large and clear LCDs are available: one on the front panel and one on the backside.
- laser diode: GaAs IR 808 nm - 1600 mW
- visable beam: 632 nm
- emission: continuous and pulsed
- duty cycle: 10% - 100%
- frequency: 10 - 20000 Hz
- laser energy: 0 - 100 Joules
- rotation of scanning plane: from -45° to +90°
- available protocols: preset and adjustable during treatment
- customized protocols: up to 80 with three steps each
- laser protection class: IV
- IEC protection class: I type BF
- dimensions: 45 x 60 x 160 cm
- weight: 60 kg